Agilent BioTek 405LS Microplate Washer, IVD
The BioTek 405 LS washer has a well earned reputation as an economical plate washer that offers excellent performance and reliability. For In Vitro Diagnostic Use.
Supplier: Agilent BioTek 405LSRVSQSI
The BioTek 405 LS washer has a well earned reputation as an economical plate washer that offers excellent performance and reliability. Washing protocols range from gentle cell washing to vigorous ELISA workflows. The 405 LS has a simple user interface for creation and execution of wash, dispense, aspirate, and maintenance protocols within the extensive onboard software
For use under computer control, Agilent BioTek LHC software used with the 405 LS is available in standard and 21 CFR Part 11 compliant versions. The optional Agilent BioTek BioStack is available for walk-away automation of up to 50 microplates.
|CE and TUV marked. RoHS compliant.|
|<3% CV: 300 μL/well (96-well washing); <4% CV: 80 μL/well (384-well washing)|
|96- and 384-well; Low profile and standard height; Solid and filter bottom (filter pore sizes 0.45 to 1.2 μm) plates|
|96-wells: 300 μL/well, 3 cycles: <30 s; 384-wells: 100 μL/well, 3 cycles: <80 s; 384-wells: 400 μL/well, 1 cycle: <20 s|
|1 to 250|
|10 to 85%, noncondensing|
|Automated Internal 4-Buffer Switching; Ultrasonic Advantage; 96-well microplate washing; Verify Technology|
|4 or 10 L (optional)|
|25 to 3000 μL/well|
|<2 μL/well (96- & 384-well plates); 96-tube manifold for 96 wells; 192-tube for 384 wells|
|Programmable in minutes and seconds, up to 60 min.|
|With internal buffer switching - 16.5 kg; Without - 13.5 kg|
|High to Low Flow; Optimized rates for cell assays|
|With internal buffer switching - 36 lb.; Without - 30 lb.|
We continue to work to improve your shopping experience and your feedback regarding this content is very important to us. Please use the form below to provide feedback related to the content on this product.
Your feedback has been submitted. Fisher Scientific is always working to improve our content for you. We appreciate your feedback.Ok